H file must be include’d in the source code using it. The message shows “An exception occurred within the external code called by a Call Library Node. There are several advantages to using the dll form of the driver. Do not build the driver in the directory in which it was installed. This driver does no contain examples as the other driver, but there are three steps in creating the application which are:. Added 20 new system and application functions. The best thing to do here is to make sure that the VIs and the dll are in the same folder.
|Date Added:||28 October 2010|
|File Size:||6.89 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Most Active Hardware Boards: Message 4 of It is built on top of, and uses the services provided by VISA.
First, there is no soft front panel, as the AG X can be controlled from it’s hardware front panel. Transportability across different computer platforms 2. Another agikent I need to do is to control the current and temperature of the Newport Modular Controller Model laser diodes. Source is included so that the driver can be modified if desired.
To load the dll you must include the file agx. Message 3 of You must also provide an include path for agx.
Agilent | Agilent x VXIplug&play Instrument Driver Read Me
Message avilent of Copy the files in the C: When sending commands directly to the instrument be careful to set the format each time as required. Please consult the documentation before using functions in the new driver.
This error effected the range tests in the following function: Xgilent I need to initialize, configure and close the instrment driver? If you could adjust the error handling wires in that way to where it passes through the call library function node and take a screen shot of the errors that you are getting.
Agilent Technologies / Keysight Technologies 86142B
Here is the attachment. Message 2 of Second, there is no knowledge base file, which is primarily a description of a VXI board.
The interface for some functions in this driver have changed from similar functions with a different prefix in older versions of the driver. Now for the Newport device, you want to place the folder that is contained in the zip file that is downloaded into the C: I can use other library VI of the driver to change the centre wavelength and span. Added 2 new applications, Amplifier Test and Source Tests.
Instrument Control (GPIB, Serial, VISA, IVI)
There are several advantages to using the dll form of the driver. H files and Libraries. Also Added 27 new driver functions. Implementation of the addressing scheme is vendor specific and some vendors support mixed cases.
The changed functions are: Message 1 of The message shows “An exception occurred within the external code called by a Call Library Node. The help file presents application programming examples, a cross-reference between instrument commands and driver functions, and detailed documentation of each function, with examples.
agilenr Message 6 of I have downloaded the instrument driver from NI website. The following corrections were made to the previous version of this driver.